Custom Search

## Monday, July 16, 2012

### Dynamic characteristic of liquid-in-glass thermometer (thermal capacitance of glass wall is included)

Its a derivation  part which is a continuation of the previous two post. Here the dynamic characteristics of liquid-in-glass thermometer by considering the thermal capacitance of glass wall is discussed.

Its better you read the previous two post before going through this, for better understanding:

Dynamic characteristic of liquid-in-glass thermometer (thermal capacitance of glass wall is included)

The heat energy balance:

(20)
Thermal capacitance of the glass walls is included.

a). The heat energy balance for mercury in the bulb:

Figure 1:. Liquid-in-glass thermometer.

(25)(26)(27)
We get the first order differential equation:

(28)

b). The heat energy balance for the glass wall has both inflow and outflow of heat:

(29)(30)
(31)(32)
(33)
Let ∆t0, then
(34)
Substitute (28) into (34) and after manipulations we get:

(35)
or
(36)
Equation (36) is a second-order differential equation.

Cg                - thermal capacitance of glass bulb, J/K;

cgp               - specific heat of glass, J/(kg*K);
Mg                - mass of glass bulb, kg;
∆Qgaccum        - amount of heat energy accumulated by glass bulb during a period of time ∆t,J;
∆Qgin            - amount of heat energy transferred to glass bulb from fluid during a period of time ∆t, J;
∆Qgout           - outflow of heat energy from glass bulb during a period of time ∆t, J;
Rf,g               - thermal resistance of fluid film and glass wall, K/W;
Rg,m              - thermal resistance of glass and mercury film, K/W;
Tg                - temperature of glass, K.

(37)
(38)

where,
Ag                - heat transfer surface area, m2;
hf,hm            - film coefficients of fluid and mercury, respectively, W/(m2*K);
kg                 - thermal conductivity of glass, W/(m*K);
xg                 - thickness of glass wall, m.

Transfer function is as follows:

(39)
where,                   ฯ1,2 = Rfl,gCm..
Let,    ฯ1 ฯ2 =ฯ2,       (39)          and    ฯ1+ ฯ2+ ฯ1,2= 2 ฮถฯ.   (40)

Then,

(41)
Let:    T'fl = A - step change, and ฮถ = 1. Then we have:

(42)
Use inverse Laplace transform:

(43)
Let:    A= 10oC and ฯ = 85,s. Then we can plot a transient response of this thermometer to a step change in the input variable (see Figure 2).

Figure 2:. Transient response of liquid-in-glass thermometer.

Article Source:: Dr. Alexander Badalyan, University of South Australia

## Labels

Automation (4) Basics (48) DCS (1) Flow (27) Flow Meters (22) Level (19) PLC (2) Pressure (12) SCADA (2) Sensors (7) Temperature (17) Transducer (17)

Hai friends…welcome to my blog. This blog is exclusively for instrumentation engineering students which will provide sources for their reference and studies. As you all know Instrumentation is now a fast emerging and developing field in Engineering. This blog has different categories like PLC, SCADA, DCS, Sensors and Transducers, Computer control of process, Industrial Instrumentation, etc.

This blog will also provide an exclusive ‘ASK ME’ section where you can make any queries and share your ideas about instrumentation. The solution for your queries will be given to you by mail from best of my knowledge and reference.

So I wish this blog may be very useful for your studies and reference.